COVID-19 Equipment Removal Process: Non-disposable Respiratory-Related Equipment
Last update: March 23, 2020
All non-disposable equipment example: Flowmeter, high flow oxygen devices, ventilators, BiPAP, video laryngoscopes, oxygen cylinders, oxygen holders, nitric oxide units, syringe pumps
Cleaning equipment will start in the patient’s room.
- Must have proper PPE during cleaning.
- Move equipment to be removed close to the patient door.
- Remove all disposable parts/pieces and discard in red biohazard bag.
- Perform hand hygiene with gloved hands.
- Clean equipment with appropriate wipes (orange top or purple top based on manufacture instructions).
- All surfaces of equipment must be cleaned
- Surface must be left to dry for manufacture required time (Purple top – 2 mins and Orange top – 4 mins)
- Equipment is passed to assistant outside room (Ante room preferred)
- Staff member inside patient room then doffs PPE and performs hand hygiene
- Doff face shield
- Exit patient’s room and perform hand hygiene
- Doff N95 respirator and perform hand hygiene
- Don gloves
- Re-wipes equipment with appropriate wipes and then allowing for appropriate surface dry time
- Doff gloves and perform hand hygiene.
- Equipment is safe to continue regular reprocessing.
These documents and content on this website are guidelines during the COVID-19 pandemic. Because new information is released rapidly, these documents can be updated or changed at any time. These documents are in no way to be considered as a standard of care and the content is not intended to be a substitute for professional medical advice, diagnosis, or treatment. The information in these documents is provided with no guarantees, accuracy, or timeliness. All content in these documents and website are for informational purposes only and do not constitute the providing of medical advice.